How do I reset my CADD pump?
To reset the reservoir volume without changing the cassette/tubing: • Make sure the pump is stopped • Press “NEXT” to display the reservoir volume screen. Press “ENTER” • If message on the screen reads RESET RESERVIOR TO ____? ML, press “Y” or the “up arrow” keypad.
How do you fix downstream occlusion?
What Should I Do If My Pump Reads “Down Occlusion”
- Check clamp on tubing and unclamp if clamped.
- Check tubing for any kinks or if patient is lying on tubing causing it to occlude.
- Make sure tubing is placed in pump correctly.
- Try to restart by pressing Pause/Yes to Resume/Run to Start.
What do you do when your IV pump is beeping?
“OCL” indicates there is an occlusion (constant beeping): To resolve, check the tubing for any kinks. Push the run/pause button to silence the alarm. Push the run/pause button again to restart the pump.
How do you prime a CADD pump?
The pump will prompt you to begin priming by asking “Prime Tubing?” Open clamp on pump tubing and press Y key. Press and hold Y key until pump tubing is fully primed. If the key is released and you are not finished priming, press and hold Y key until priming is complete.
How do you fix upstream occlusion?
To help prevent upstream occlusions, it is important to completely spike the IV container, remove the blue slide clamp completely from the keyhole, disengage the blue slide clamp completely from the IV tubing, check that the IV tubing is clear of any kinks or collapsed sections, ensure the roller clamp (if present) is …
What causes downstream occlusion?
When an IV becomes obstructed, the pressure within the IV syringe and connected tubing will increase. When the pressure reaches its setting threshold, a downstream occlusion alarm will be activated.
What can cause an upstream occlusion?
An upstream occlusion refers to a blockage that occurs before a fluid has reached the pump, commonly caused by improper connections from an IV bag to the pump, a kink or collapse in IV tubing, issues with tubing clamps and more.
Why does the IV machine keep beeping?
Your pump may alarm or “beep” for a number of reasons. It will alarm if a medicine is finished infusing or if the pump detects a blockage or air in the line. Always call your nurse when the pump starts to alarm.
What causes an IV pump to alarm?
There is a large variety of infusion alarms that may set off for technical reasons (e.g., door open, infusion completed) or infusion problems (e.g., occlusion, air- in-line detection). An incidence of 2.2 infusion alarms per patient per day has been observed in 5 ICUs.
How do you prime a CADD PCA pump?
What causes upstream occlusion?
What is pump occlusion?
What does occlusion mean? Occlusion is the squeezing of peristaltic pump tubing between the pump head rollers and the tubing bed.
What is a downstream occlusion?
One of the many important features of an infusion pump is the “Downstream Occlusion Alarm”. When an IV becomes obstructed, the pressure within the IV syringe and connected tubing will increase. When the pressure reaches its setting threshold, a downstream occlusion alarm will be activated.
What causes IV pump to alarm with occlusion downstream?
How do I change the lock level on a CADD pump?
- Hope Healthcare Draft protocol for CADD – Legacy Model 6300 pump – January 2005. LOCK LEVELS.
- TO CHANGE THE LOCK LEVEL. Press LOCK.
- Press LOCK. Code 0 will be displayed.
- Press the increase button till “63” appears. Press LOCK to set to set the new lock level.
How do I test the CADD®-Solis VIP pump?
•Become familiar with the CADD®-Solis VIP pump before performing the following tests and procedures. Read the Operator’s Manual supplied with the pump before proceeding. •Run the test procedures in manual mode. •Many of the tests include steps to unlock the keypad. This is necessary only if the keypad is not already unlocked.
Can patients have access to the CADD-legacy®PCA pump?
Do not permit patients to have access to this card, as the information would allow access to all programming and operating functions. Assistance with the CADD-Legacy®PCA pump is available to clinicians 24-hours-a-day by calling 1-800-258-5361.
Is there a technical manual for the CADD 2120 VIP ambulatory infusion pump?
This Technical Manual is applicable only to the model 2120 CADD®-Solis VIP ambulatory infusion pump. It is intended to provide a basic but limited understanding of the mechanical and electrical operation of the CADD®-Solis VIP ambulatory infusion pump to people familiar with the device.
Is there a technical manual for the CADD Solis VIP infusion pump?
Introduction This Technical Manual is applicable only to the model 2120 CADD®-Solis VIP ambulatory infusion pump. It is intended to provide a basic but limited understanding of the mechanical and electrical operation of the CADD®-Solis VIP ambulatory infusion pump to people familiar with the device.